- New Technologies
- Open Access
Intraprocedural guidance for recanalization of post-thrombotic venous lesions using live overlay of center lines from pre-operative cross-sectional imaging: a preliminary experience
CVIR Endovascular volume 3, Article number: 32 (2020)
Pre-procedural contrast-enhanced CT and MRI imaging is typically acquired prior to deep venous recanalization procedures for post-thrombotic syndrome. This technical note reports the utility of live-overlay of augmented centerlines extracted from pre-procedural CT and MRI imaging in facilitating fluoroscopic-guided recanalization of post-thrombotic venous lesions.
Methods and materials
Six patients with pre-procedural CT or MR venography data were incorporated into a commercially available 3D overlay software (Vessel Assist, GE Healthcare, Buc, France) during venous disease interventions for post-thrombotic venous lesions. Procedures were performed on the GE Discovery IGS 740 fluoroscopy system. After manual determination of the vasculature from preprocedural CT or MR, centerlines were created representing the location and trajectory of the vessels. Steps showcasing the creation of centerlines and their representation during overlay with real-time fluoroscopic guidance in these cases are outlined. Time required to cross the post-thrombotic and occlusive venous segments were reviewed.
All iliocaval recanalization procedures were successfully performed utilizing vessel centerline 3D overlay. In one case where occlusion extended to the femoral vein, mis-registration was identified over the femoral anatomy due to a complex leg rotation compared to pre-procedural imaging. No procedural complications related to utilization of software were noted. Average crossing time for occlusions was 3.4 min (range 1.6–5.2).
3D overlay with vessel tracking from pre-procedural CT and MRI imaging is technically feasible and assists in catheter navigation for post-thrombotic venous segments. While results from these preliminary experiences support the continued use of this technology, further prospective and comparative evaluation of this technique is warranted to assess for added value in technical success, reductions in procedure time or reductions in radiation exposure.
Recanalization of chronic deep venous obstructions or post-thrombotic venous segments can require extensive fluoroscopic time, contrast use, and repeat interventions (Barbati et al. 2019). One contributing factor to the challenge of such cases is navigating the occluded segment rather than collaterals coursing in a similar trajectory, especially when considering sharp recanalization techniques (Cohen et al. 2018). Contrast-based road mapping for device navigation can be variable in delineating occluded vein segments or patent segments leading to normal venous outflow (McDevitt et al. 2019). As such, strategies to improve operator confidence during times of catheter guidance are invaluable in achieving procedural completion and technical success. Augmented reality (AR) has been shown to improve procedure performance in other challenging areas such as neurosurgery (Contreras Lopez et al. 2019), by projecting representation of structures of interest extracted from the pre-operative exams on live imaging during the procedures. Cases using an augmented vasculature model for venous recanalization have been reported previously (Chinnadurai and Bismuth 2018). In this patient cohort, poor contrast opacification due to chronic post-thrombotic changes prevented adequate guidance for catheterization. Trajectories referred to as “centerlines” were created by using contrast-enhanced CT and MR imaging acquired prior to recanalization. These centerlines corresponding to the location of the post-thrombotic and often hypoplastic venous segment, as delineated by the interventional radiologist, can be superimposed on live fluoroscopy to improve vessel catheterization. This technical report highlights the feasibility of utilizing centerline fluoroscopy overlay from pre-operative CT and MR imaging during traversal of post-thrombotic venous lesions.
Materials & methods
Following IRB approval, a retrospective review was performed to identify cases in which centerlines from pre-operative imaging were incorporated during venous recanalization. From July 2016 (when software was made available to the authors’ institution) to December 2017, six consecutive patients with post-thrombotic iliocaval venous lesions were identified. Low-sample size stemmed from software implementation during hours of operation in which support staff was available in the event troubleshooting was required. Five patients had pre-procedural contrast-enhanced CTs while one patient had pre-procedural contrast enhanced MRI. Patients were treated on the same angiography unit (Discovery IGS 740, GE Healthcare). Before the start of each procedure, cross-sectional images from the most recent pre-operative scans were loaded from PACS to a designated workstation (Advantage Workstation, GE Healthcare). Each dataset was prepared with one segmented volume to be used for initial registration and vessel centerline creation for the vessels of interest.
The technique for centerline creation, vessel tracking, and fluoroscopic overlay is shown in Fig. 1. To initiate the creation of the vessel centerline, multiple points of interest were manually entered in the axial projection from the CT or MR imaging. A final point was positioned to indicate the most central part of the vessel to be tracked. A preliminary template of the centerline created from those points was outlined by the software on each cross-sectional view. The user then refined the centerline curvature as needed using a dedicated tool (Fig. 1a-g and 2).
In cases of multivessel recanalization, the method was repeated in similar fashion on the other vessel(s). With pre-operative CT datasets, bones were used for initial registration to live fluoroscopy and were automatically segmented by the software. Pre-operative MRI datasets required manual segmentation of the femoral heads or other structure of interest to allow for registration to live fluoroscopy (Fig. 3). In patients with pre-existing stents (if present in a different vascular segment than the one requiring recanalization), initial registration could be supplemented with these fixed reference points rather than the bones alone (Fig. 4).
Once the patient was on the angiography table, and before the procedure started, the segmented anatomy and the vessel centerlines from the pre-operative scan were loaded into the overlay software and registered to the patient by the technologist using two fluoroscopic images 600 apart (Fig. 1h-i). The overlay display was made available to the operators on the large display monitor, side-by-side with the native fluoroscopic images (Fig. 1j-k). Refinement of the registration was performed if the operators noticed mis-registration. Digital zoom was also used with the overlay display to reduce the need of field-of-view magnification related exposed dose. Stored fluoroscopic images of each case were retrieved and time information reviewed, to compute the occlusion crossing times.
Table 1 summarizes the six patients, their presentation pathologies, procedural parameters, and clinical follow-up. Technologists performed the overlay setup in less than 5 min in all cases. No errors or difficulty in setup were reported. The physicians for all iliocaval cases considered accuracy of the registration appropriate. In one patient, where the occlusion extended to the femoral vein, mis-registration was identified over the femoral anatomy due to differential leg rotation compared to CT. This was corrected with manual translation and rotation readjustments using osseous segmentation as a landmark. No procedural complications related to software utilization were recorded. Time to cross the post-thrombotic venous segments with a hydrophilic 0.035″ wire was on average 3.4 min, ranging from 1.6 min to 5.2 min among all cases. In one patient, no contrast was utilized to visual the post-thrombotic venous segment during intra-procedural catheterization due to the presence of the centerline guidance.
While the benefits of fluoroscopic 3D overlay created from procedural digital subtraction angiography and cone-beam CT acquisitions are well demonstrated, the utility of incorporating peri-operatively acquired imaging data during vascular interventions remains limited (Gorges et al. 2006; Bapst et al. 2016; Tacher et al. 2017). This report showcases the successful use of 3D overlay of centerlines from pre-operative imaging as roadmaps during venous recanalization interventions in six different patients.
Contrast-enhanced CT and MR imaging was sufficient for the provider to identify the course of the post-thrombotic venous segments and allowing for creation of a centerline. The software automatically calculates trajectory based on luminal opacification of the proximal and distal ends in the contrast enhanced cross-sectional examination with registration of the non-opacified portions of the vessel performed by bridging segments with manual clicks by either the operator or technologist. Such automatic registration does not correlate well with vessel wall thickness due to poor wall visibility relative to intraluminal post-thrombotic changes limiting the software’s ability to display a theoretical outline of the vessel.
Currently, the implemented centerline technology relies on physician determination of the affected post-thrombotic venous channel. In the future, it may be possible to utilize this technology with non-contrast imaging based on landmark selection and triangulation with a selected vascular tract, though this was not formally interrogated in this study. Further automated segmentation of anatomical landmarks from imaging datasets for registration with fluoroscopic images may also be considered in the future.
The use of augmented centerlines represents a promising road mapping tool regardless of the position of the C-arm. Displaying vessel centerlines allows for guidance through long segments of poor vessel opacification of contrast due to post-thrombotic changes. In the future, an augmented reality environment could be overlaid to the patient using more advanced visualization tools such as virtual reality glasses.
Limitations of this technical note include a small retrospective patient cohort and lack of a control group showing differential time to recanalization or contrast volume use. Given the superior spatial resolution and standardization of dose parameters of CT compared to MRI, CT is preferred in the identification, extrapolation, and standardization of centerlines from occluded and atretic vascular channels over MR. However, in patients with prior MR imaging for other indications, these can be alternative options to develop guidance overlay centerlines. In one patient with pre-existing right external iliac to femoral vein stents and a left iliocaval post-thrombotic occlusion, a crossing time of 1.6 min was recorded, which is lower than others in this cohort. Presumably, the patient’s preexisting stents provided a more accurate landmark than the adjacent pelvic osseous structures when affixing 3D centerlines to the live images.
In conclusion, 3D overlay with vessel tracking from pre-procedural CT and MR imaging data for catheter navigation during venous recanalization interventions is feasible. Procedure time to cross the occlusion was on average 3.4 min in this particular cohort, noting that reducing crossing times subsequently reduces procedure length in these often time-intensive procedures. Specifically, through a combination of the centerline creation and overlay process, intravascular ultrasound incorporation, and understanding of catheter and wire locations based on fluoroscopic landmarks, the potential for reductions in procedural contrast accumulation and dose exposure without compromising safety exists and would be fodder for future investigation in a prospective nature.
Availability of data and materials
All data reported in this manuscript is available for review if requested. All patient related data recorded for the purposes of this manuscript is stored in a secure HIPAA-compliant data storage repository.
Bapst B, Lagadec M, Breguet R, Vilgrain V, Ronot M (2016) Cone beam computed tomography (CBCT) in the field of interventional oncology of the liver. Cardiovasc Intervent Radiol 39(1):8–20
Barbati ME, Gombert A, Schleimer K et al (2019) Assessing radiation exposure to patients during endovascular treatment of chronic venous obstruction. J Vasc Surg 7(3):392–398
Chinnadurai P, Bismuth J (2018) Intraoperative imaging and image fusion for venous interventions. Methodist Debakey Cardiovasc J 14(3):200–207
Cohen EI, Beck C, Garcia J et al (2018) Success rate and complications of sharp recanalization for treatment of central venous occlusions. Cardiovasc Intervent Radiol 41(1):73–79
Contreras Lopez WO, Navarro PA, Crispin S (2019) Intraoperative clinical application of augmented reality in neurosurgery: a systematic review. Clin Neurol Neurosurg 177:6–11
Gorges S, Kerrien E, Berger M-O et al (2006) 3D augmented fluoroscopy in interventional neuroradiology: precision assessment and first evaluation on clinical casesPaper presented at: workshop on augmented environments for medical imaging and computer-aided surgery - AMI-ARCS 2006 (held in conjunction with MICCAI’06); 2006-11-06, Copenhagen
McDevitt JL, Srinivasa RN, Gemmete JJ et al (2019) Approach, technical success, complications, and stent patency of sharp recanalization for the treatment of chronic venous occlusive disease: experience in 123 patients. Cardiovasc Intervent Radiol 42(2):205–212
Tacher V, Petit A, Derbel H et al (2017) Three-dimensional image fusion guidance for Transjugular intrahepatic portosystemic shunt placement. Cardiovasc Intervent Radiol 40(11):1732–1739
We thank Pinar Ozbek from GE Healthcare for her contribution and insight in the review of conceptual theory and appropriate clinical applications of the Vessel Assist software.
No sources of funding were necessary for the study and collection, analysis, interpretation of data, and writing of this manuscript.
Ethics approval and consent to participate
Data regarding subjects included in this manuscript were reviewed and reported following institutional IRB approval. All patients reported in this manuscript were consented for permission of data inclusion for research purposes prior to performing the stated interventions.
Consent for publication
All patients reported in this manuscript were consented for publication of procedural data for research purposes prior to performing the stated interventions.
Please note that Raphael Doustaly and Gregoire Avignon are employees of GE Healthcare and David C. Madoff is a paid consultant for GE Healthcare. The authors otherwise have no competing interests or conflicts of interest to disclose.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Sundararajan, S.H., Doustaly, R., Avignon, G. et al. Intraprocedural guidance for recanalization of post-thrombotic venous lesions using live overlay of center lines from pre-operative cross-sectional imaging: a preliminary experience. CVIR Endovasc 3, 32 (2020). https://doi.org/10.1186/s42155-020-00121-6
- Vessel overlay
- Vessel tracking
- Vessel ASSIST
- Pre-procedural CT MR venography
- Fluoroscopic live overlay
- Chronic venous occlusion
- Iliocaval thrombosis
- Venous recanalization